## 1.5 V Inductance Meter Circuit

**Designed and Presented by: Abu-Hafss**

The circuit is pretty straight forward wherein, two NPN transistors are cross-coupled to form a flip-flop oscillator. The values of R1 and R2 could be anything between 47 – 100R. The frequency of the oscillation is inversely proportional to the inductance and it can be calculated with the following formula:

Frequency (kHz) = 50,000 / Inductance (uH)

CALIBRATION:

Initially the circuit has to be calibrated using a known inductor, as described below:

Suppose, we have an inductor of 100uH. Putting the value of inductor (100uH) in the above formula, we get 500kHz.

Connect the inductor cross point A & B and power on the circuit. It will start oscillating.

Connect the frequency meter at point A or B and ground.

Adjust the POT until the meter reads 500kHz. Now the circuit is calibrated.

INDUCTANCE MEASUREMENT:

Connect an unknown inductor across A & B.

Power on the circuit and read the frequency at point A or B.

The above formula can also be written as:

Inductance (uH) = 50, 000 / Frequency (kHz)

Putting the value of frequency in this formula, the value of the inductor can be found.

Waveform Image:

Hi Abu Hafss

This looks so simple its to good to be true,I'm going to try it.

Where did you derive that formulae from.

Regards Robin

Hi Robin

Here is the waveform of the oscillation

https://dl.dropboxusercontent.com/u/20969135/Waveform%20Ind%20Meter.png

The frequency of oscillation depends on the L/R time constant comprising the inductor and resistors R1 and R2. The time the waveform takes to change its state is directly proportional to the inductance. In the waveform we see, the time taken for one complete cycle is 2µs for a 100µH inductor.

In other words, inductance (100µH) = Time period (2µs) x 50 ……….(A)

And we know, Frequency = 1 / Time Period OR Time Period = 1 / Frequency

So replacing Time Period in Formula (A), we get

Inductance (µH)

= 50 x (1 / Frequency) OR

= 50 / Frequency (Hz) OR

= 50 x 1000 / Frequency (Hz) x 1000 OR

= 50,000 / Frequency (kHz)

Please note, that this circuit is quite sensitive to the supply voltage and it is fairly good to about 10µH.

It is good idea to a have a known value inductor say 100µH fixed in the circuit for calibration purpose. That calibration inductor and the inductor under test may be toggled using a SPDT switch.

Hi Swagatum

I now understand it fully,I built the circuit and calibrated it and tested 3 other known inductance's ie:47uH and got 53.5uH,4.7mH and got 5.9mH,3.9mH and got 4.9mH.

Those figures will give me a good idea of what I've wound.

I also built the other circuit with the 100uA meter which may give more accurate results?But have no idea on how to fault find it and split it into smaller sections so as to check that you are on the right track

Just getting the above results from that tiny little circuit was a big boost for me

Thanks Robin

Thanks for testing the circuits!

The 555 based design will need different sets of "C" for producing the corresponding sets of measuring range, however the design will not give accurate results with lower capacitance and may show vibrating or oscillating meter needle.

Hi Robin

Please note that each time the circuit is powered on, it does need to be calibrated. Otherwise, we might get variation in readings.

Hi Abu Hafss,

Its a simplest one…thanks for sharing.

I have some question

1. Does the calibration depends on cell voltage.

2. If so then can i give a fixed 5v supply to this circuit.

Hi Vishal Sutar

1. Yes.

2. Yes.

3. Please take note of my post of Nov. 5, 2014.